A new and effective way of completing your root canal therapy utilizes a unique laser system that goes beyond what traditional root canals can offer. It's called Photon Induced Photo-acoustic Streaming (PIPS), and we are very excited to offer such a high quality service to our patients. PIPS was invented by a general dentists, Mark Colonna DDS, using the Lares Research PowerLase AT Er:YAG/Nd:YAG laser in one of his operatories about 10:30pm on a Saturday night. This process is so unique that most general dentists and endodontists do not understand the basics of how it works. They have developed tunnel vision thinking that the laser fiber is inserted into the root canal and pulled about burning up anything in its path. This could not be further from the truth. As the name implies, PIPS doesn't burn up anything, but instead transfers the laser pulse energy into a small controlled explosion of water molecules that sends a shock wave down the root canals to KILL ALL the bacteria and to remove ALL the debris/smear layer inside the root canal's neuro-vascular system. Yes, I know this statement will disturb many of you in the "Endo World", but don't get all bent out of shape until you follow some of the current studies that are going to be published soon.
ACCESSThe process of doing a root canal with PIPS starts off just like every other root canal with a well designed access into the pulp chamber so that all the canals are easily instrumented, and of course, a rubber dam.
INSTRUMENTATIONBefore putting any instrument into the canals I usually PIPS over the orifice of each canal for at least 20 seconds. I did this on an upper second molar just the other day that I was having a hard time negotiating the canals because of pulp stones. When I was done all the pulp stones were gone and I had a clear pathway into each canal. The Er:YAG laser R12 hand piece was developed to accept different laser tips made of different materials from saphire to quartz. The PIPS tip is a quartz fiber 400? in diameter that has a cone (radial) shaped tip. This cone shape allows the laser pulse to emerge from the end in a radial pattern that sends the acoustic stream of shock waves throughout the entire root structure. This is achieved by placing the Er:YAG PIPS tip into the chamber over each canal sequentially and streaming out the debris from inside the root system for 20 seconds at a time. The dental assistant flows a continuous stream of water into the chamber to allow the laser to explode the water molecules. Next, File-Ease from Ultradent is introduced to the canals via a Navi-tip. A #6 file is used to find the working length of all the canals using an electric apex locator. The canals are PIPS'd again with a water stream and the process is repeated for the next file sizes up to a#20 or 25 file size. Once this is achieved, the canals are PIPS'd with EDTA then water four times 20 seconds. This process completely sterilizes and removes all the debris and smear layer from the entire root canal system.
OBTURATIONThe way we obturate a PIPS endo case is simple. I first use a small paper point to dry up some (but not all) of the water left inside the canals. Using a measured (1 mm short of the apex) #20 Navi-tip from Ultradent, EndoRez (Ultradent) sealer is injected into each canal. This is the cool part. While injecting the sealer into the canals you can see the sealer escape out many accessory canals in the floor of the chamber. This sealer is extremely hydrophilic and it follows water throughout the entire tooth's neurovascular system. Once the canals are full, a resin coated gutta percha point (Ultradent) is placed into each of the larger instrumented canals (no condensation) and seared off with a touch and heat. Cavit or purple composite flowable resin is then placed to seal off the chamber from oxygen and micro-leakage. Occlusal reduction finishes up the root canal treatment. Sometimes the EndoRez will "Poof" out the apex, but this is usually OK because the body absorbs it rapidly. As I understand it, this sealer is based on an orthopedic cement used to fixate artificial joints and is very compatible with the bone. I actually have call this a "bone magnet", but I have been told that is irresponsible and not substantiated in the literature by my Stonehenge "teacher" whom I appreciate more than he knows, so I retract this statement. Notice the MB2 accessory canal going to the pulp chamber. You can actually see these being back-filled as you inject the EndoRez sealer all over the chamber floor.
Why is this technology important?Dentistry in general has been moving into a new era of minimally invasive treatment. The veneers are getting thinner, cavity preparations are getting smaller, and root canals are also becoming less invasive. As a matter of fact, I will predict that in the future root canals may be done without any instruments being placed inside the the root canal neurovascular system at all. Calcified canals, internal resorption, and extremely curved canal will not pose the problems they currently do today with other systems. The time it takes to do a root canal will shorten and more and more general dentist will find it easier to do endodontics.
Can any dentist with a Er:YAG laser do a PIPS root canal?No. So far only those dentists with proper training and the Lares Research PowerLase AT, SPA or LightWalker can do a PIPS root canal. Below are some of our more unusual case which are becoming more usual everyday. Check out these case and try to count the accessory canals for each root. Tough case. Bridge abutment with irreversable pulpitis and calcified canals. This is where PIPS really shines. The canals opened and instrumented to #25 file size. "More information on PIPS Endo with the latest Lares Er:YAG lasers".
PIPS "Tips":This section will be devoted to some of the little problems I've noticed during this procedure:
- Before doing any PIPS procedure with your new laser, take a course!!!
- I always use a bite block when using this technique. We were working on #22 when suddenly, without notice the patient bit down. The PIPS tip when down the canal and broke right off! It was wedged in as snug as a bug in a rug. Could not get anything around it. Luckily, the prep wasn't as wide as the tip all the way down to the apex so I took a long fine pointed gold diamond and chewed it up from the orifice, and then PIPS'd it out.
- Don't put the tip too deep inside the orifice. I've seen that this limits the "PIPS Effect" of rapidly moving fluid in and out of the canal. You can regulate the PIPS effect by moving the tip up and down until you get the churning of the debris out of the canal.
- It is easier to see the PIPS effect if you don't use a heavy stream of water into the chamber. You need just enough to keep the chamber full and to replace the dirty liquid with clean. This speeds things up by helping you know when you can move to the next step, because your water is clear and no more junk is coming out of the canals.
- Don't touch the tip of the laser PIPS tip to the inside of the chamber. This will wear the point off rapidly!
- Don't run the PIPS tip inside the tooth dry. It will not hurt the tooth but it will "abrade" off your PIPS tip quicker with the back splatter.
- Try to open the canals wide enough to get the appropriate NaviTip to within 2mm on the apex. Make sure you test the back pressure needed to express a tiny amount of EndoRez out the tip before placing it into the canal. This will help so you will not shoot a lot of sealer out the apex. Then just barely inject the sealer. Pull the NaviTip back until it is no longer tight in the canal and slowly inject until you see it coming out the orifices.
- The "Perio radial PIPS" tip are thicker and stronger than the Endo PIPS tips and in my opinion work just as well and the endo tips. And they are about $100 less expensive! However, like any other new system, it would be wise to follow the instruction and use the regular PIPS tips for endo, or you will be outside the guidelines set forth by the manufacture, and researchers. They have just come out with a new perio tip that is not as effect as the old because it covers too much of the tip with a sleeving to protect the tip from calculus trauma during WPT. However, I am not so sure the radial tip is that important to the overall PIPS performance because we have done some very good treatments with the point burnt off of our PIPS tips (A good research project for an Endo Grad Resident?)
- Vital or hyperemic pulps are a little harder for the PIPS effect to dislodge than necrotic tissue. Using NaClO will help a lot. Since most of my final endodontic procedures first start off as toothaches, we usually do a pulpectomy first. I do this with files from #6-15, Gates Glidden (2-3), PIPS with peridex (1 minute in each canal or until no more" junk" comes out of the canals), place calcium hydroxide paste (using #15 NaviTip), cotton, IRM, occlusal reduction. This makes the pulp tissue very easy to remove with PIPS/EDTA during the final endodontic procedure.
- If while injecting the EndoRez you see sealer coming out of many other instrumented orifices), don't assume you need not inject into them independently. There maybe anastomosis that need to be seal through each and every instrumented canal that would be missed otherwise.
- No one is telling you you cannot use your own long tried and true method of doing endodontics with the PIPS technique. You can use it as an adjunct to any existing technique.
- Cleaning the PIPS tip is difficult. We spray it down with viral/bacterial disinfectant and then let it sit wet for several minutes (until the next procedure). However, this maybe a good ways to prevent cross contamination, it is not always the best for your laser. Pay very close attention to how wet with disinfectant the base of the laser tip that is inserted into the R-14 handpiece. If it is wet AT ALL (even damp), then when you fire the Er:YAG it will hit the moisture and splatter it back onto the handpeice's mirror. After a short while you will notice the laser being less and less powerful to a point it no longer works. Changing the mirror is the only remedy.
- Because the laser arm is balanced to hold its position when you let go of it, it can sometimes cause a problem. I have more than once pushed the arm to the side without paying attention and as I pulled my hand off of the hand piece my glove caught the laser fiber and broke it in half. To avoid doing this you should always act like a robot and watch your hand place the arm/handpiece far enough away from you so this costly mistake doesn't happen to you too.
UPDATE 11/2011As expected with any new procedure, updates will trickle in from time to time. Recently, my oldest son Dr. Phillip Kraver when to get his certification in both PIPS and WPT. Of course I sat down and picked his brain about what they we're currently teaching and came to the conclusion that some parts of the procedure had been changed. No game changing information, but different none the less. They now have a Lightwalker Facebook group to discuss all of the procedures so much more information will be able to be found in those discussions: Lightwalker Facebook User Group
- After the initial instrumentation to find the patent canals 5.25% sodium hypochlorite (NaOCl) is PIPS'd into the canals for 20 seconds and then let set through the next size file instrumentation. This is in response to a recent study that will be published that showed 100% killing of E. facalis at the apex of the root canal's neural vascular system. I previously reserved NaOCl irrigation for canals with hyperemic tissue still present because of its digestive properties. This is because NaOCl interferes with the set of EndoRez sealer that I use in the obturation of the root canal.
- Now they are saying that the PIPS affect is so strong that it is unnecessary to place the PIPS tip directly in or even over each canal. Just placing it into the access and this is apparently enough. This is good news since negotiating the canals with the tip is one of the biggest cause of breakage of the PIPS tips.